1. Enter the serial number on the pipette handle in the serial number column, select the volume range of the pipette to be calibrated from the drop-down list, and also select the room temperature. The test volume (µL), accuracy (%) and precision (%) of the pipette to be calibrated will appear automatically on the screen.
2. Please enter the results of 5 consecutive weights (in mg) in the below columns numbered 1-5.
3. Click the “submit” button to get the test results. Accuracy and precision should be within their standard ranges.